If you want to perform an extremely accurate Biofeedback scan, baseline is necessary. Baseline could disregard electrical noises, Wi-Fi signals and any other issues that may affect your scan and make your scan results even more accurate.
You are recommended to stick the two TENS pads together during baseline if you want to do a biofeedback scan with TENS Pads. However, John White recently stumbled upon that your scan result will be more accurate if you connect the TENS pads to your body during baseline.
An Accidental Discovery
During the weekend, John performed biofeedback scans on a couple of people.
The first person has deteriorating health. In recent years, his ability to balance and talk fluently has declined. Doctors had told him that his brain is lacking in oxygen.
John did a baseline first, but didn’t connect the TENS pads together as we recommend. Instead, he did the baseline with the TENS pads attached across his stomach, using the GX General Biofeedback Scan (C) – JW preset. A few minutes into the baseline, John realized his error, but decided to continue anyway, to see what happens. After the baseline was completed, he did the scan without moving the TENS pads.
The first thing John noticed was that there were not many hits. No surprise there. But the few hits reported were right on the money.
Then John did a Reverse Lookup. According to the report, “Hypoxia Brain” is exactly what this person has!
Thinking this might be something important, John scanned the second person using the same procedure. This person is a heavy smoker, and is currently feeling under the weather.
This result was scary because it found the Royal Rife frequency for severe disease. And this person smokes about a packet a day.
We know that Reverse Lookup is not a diagnostic tool, but it would appear that this new method of doing a biofeedback scan improves the accuracy of the results by an order of magnitude.
How to Do Biofeedback Scan Using This Method
It is important to perform a scan as quickly as possible after the creation of a baseline. And with this new method, the TENS pads are not moved.
So you can sit and have a baseline, and then click Scan button immediately after doing the baseline. Try to keep your TENS Pads as still as possible.
With this in mind, John added a new checkbox to allow this.
Download the latest software Spooky2 20191123 here: https://www.spooky2.com/downloadspage/
Place two TENS Pads where you want to scan.
Open your software.
Go to the Presets tab, click >Biofeedback>GeneratorX and choose the preset you want.
Go to the Control tab, click “Overwrite Generator” and choose your generator for a scan.
Tick Baseline Before BFB, and then click the Scan button in the biofeedback scan panel.
Now, a biofeedback scan will be automatically performed when you finished the baseline.
Let’s try the new scan method now! Comment below and share your scan experience with us.